### arXiv papers

Download open scientific papers. Check out the SCG's arXiv papers

# Spatial and frequency domain techniques for segmentation of Placido images and accuracy implications for videokeratography

## LAV de Carvalho and OM Bruno

### COMPUTER METHODS AND PROGRAMS IN BIOMEDICINE, 79(2):111-119, 2005

Objective: Videokeratography (VK) has been a widespread technology for corneal surface analysis since the mid 1980s. Most manufactures use personal computers attached to a Placido disc apparatus in order to capture and process digital images. Although precision reported by most manufactures are within very good limits, none of them have disclosed, probably due to proprietary reasons, the nature of the algorithm used in their image-processing phase. This is a problem when researchers want to reproduce or test their own curvature or elevation algorithms on Placido images generated on different commercial videokeratographs or even compare their algorithms on data from different manufactures. Our main objective in this work was to develop certain basic techniques for Placido, image edge detection and to compare the results of each algorithm in terms of precision at the image level and also the consequences for axial curvature computations. We also propose that manufactures come forward and at least explain which image-processing technique is used in their own algorithms so other researchers and laboratories can make better use of their data to improve VK algorithms. Material/methods: Placido images from an Eyesys system 2000 were captured for four different spherical surfaces. Each image was saved in bitmap format at the hard disk of an IBM computer. Six different image-processing algorithms were developed using different techniques well documented in the literature. The six methods were as follows: (1) first order numerical derivative, (2) first and (3) second order Fourier derivative, (4) the Marr-Hildreth filter, (5) Canny's method and (6) Mathematical Morphology. Each algorithm was tested on each of the Placido images. Results: Edge radial distance from center of Placido image was compared for each algorithm and a computer simulation of the VK system. The simulated image was used as absolute reference. Another approach was to calculate Axial dioptric power using, again, well documented procedures, and compare the results for each image detection algorithm. Mean deviation in terms of pixels/millimeters/dioptric power for all spheres for methods (1-6) were, respectively, (1) 33.1695/0.7961/0.79, (2) 32.79/0.7870/0.7724, (3) 60.7150/1.4572/1.4192, (4)18.97/0.4553/0.4572, (5) 46.33/1.1119/1.0917 and (6) 20.55/0.4932/0.48. Discussion: All methods have great deviation propagation in terms of dioptric power calculations when the axial algorithm is used and the absolute reference simulated edges are used to generate the calibration curves. This indicates that researchers should be more careful when using resulting image processing files from different videokeratographs to compare their own curvature or elevation algorithms among different instruments or even to measure the absolute precision of their new algorithms. (c) 2005 Elsevier Ireland Ltd. All rights reserved.